No Access Submitted: 08 July 1999 Accepted: 05 October 1999 Published Online: 28 December 1999
Review of Scientific Instruments 71, 168 (2000); https://doi.org/10.1063/1.1150179
more...View Contributors
• G. M. B. Parkes
• G. Bond
• P. A. Barnes
• E. L. Charsley
This article describes the design and operation of a new thermal analysis instrument which uses microwaves to simultaneously heat and detect thermally induced transformations in samples with masses in the range of 50 mg to 0.5 g. The data acquisition and control software developed for the instrument support a range of experimental techniques including constant power, linearly ramped power, linearly ramped temperature, and various modulated methods. Microwave thermal analysis utilizes the fact that physical or chemical alterations in a material, caused by processes such as melting, decomposition, or solid-solid phase changes, cause variations in its dielectric properties. These can be revealed by a variety of means including changes in the sample temperature, the differential temperature, or the shape of the power profile during linear heating experiments. The scope of the instrument is demonstrated with the decomposition of basic copper carbonate. The large temperature increase $(∼100 °C)$ observed on the formation of the strongly coupling oxide indicates the potential sensitivity of using the thermal effects of dielectric changes as a means of detection. Further fine detail can be revealed by the use of derivative plots of either the applied power or the temperature.
1. 1. E. Karmazsin, R. Barhoumi, and P. Satre, J. Therm. Anal. 29, 1269 (1984). Google Scholar
2. 2. A. C. Metaxas and R. J. Meredith, Industrial Microwave Heating (Peter Perigrinus, London, 1983). Google Scholar
3. 3. D. M. P. Mingosand D. R. Baghurst, Chem. Soc. Rev. 20, 1 (1991). Google ScholarCrossref
4. 4. J. Dolandeand A. Datta, J. Microwave Power Electromagn. Energy 28, 58 (1993). Google ScholarCrossref
5. 5. H. F. Huang, J. Microwave Power 11, 305 (1976). Google Scholar
6. 6. S. Evansand J. Penfold, J. Microwave Power Electromagn. Energy 28, 84 (1993). Google ScholarCrossref
7. 7. G. M. B. Parkes, P. A. Barnes, E. L. Charsley, and G. Bond, J. Therm. Anal. 56, 723 (1999). Google ScholarCrossref
8. 8. M. Ravindran, P. Monsef-Mirzai, J. K. Maund, W. R. McWhinnie, and P. Burchill, J. Therm. Anal. 44, 25 (1995). Google ScholarCrossref
9. 9. G. Bond, R. B. Moyes, S. A. Pollington, and D. A. Whan, Meas. Sci. Technol. 2, 571 (1991). Google ScholarCrossref
10. 10. E. Karmazsin, R. Barhoumi, P. Satre, and F. Gaillard, J. Therm. Anal. 30, 43 (1985). Google ScholarCrossref
11. 11. J. G. Zieglerand N. B. Nichols, J. Dyn. Syst., Meas., Control 115, 220 (1993). Google ScholarCrossref
12. 12. G. M. B. Parkes, P. A. Barnes, E. L. Charsley, and G. Bond (unpublished). Google Scholar
13. 13. G. M. B. Parkes, P. A. Barnes, E. L. Charsley, and G. Bond, Anal. Chem. (in press). Google Scholar
14. 14. T. Labuzaand J. Meister, J. Microwave Power Electromagn. Energy 27, 205 (1992). Google ScholarCrossref
15. 15. J. G. Binner, J. A. Fernie, and P. A. Whitaker, J. Mater. Sci. 33, 3009 (1998). Google ScholarCrossref
16. 16. B. G. Ravi, P. D. Ramesh, N. Gupta, and R. J. Rao, J. Mater. Chem. 7, 2043 (1997). Google ScholarCrossref
17. 17. N. N. Greenwood and A. Earnshaw, Chemistry of the Elements, 2nd ed. (Butterworth-Heinemann, Washington, DC, 1997). Google Scholar
18. 18. P. Monsef-Mirzai, M. Ravindran, R. McWhinnie, and P. Burchill, Fuel 74, 20 (1995). Google ScholarCrossref
19. 19. Z. D. Zivkovic, D. F. Bogosavljevic, and V. D. Zlalovic, Thermochim. Acta 18, 310 (1977). Google ScholarCrossref
1. © 2000 American Institute of Physics.